Accutrend GC Blood Pressure Monitor pdf manual download. Since each code strip and test Glucose ® Accutrend, GC, GCT, DM CODE 50 TESTS. Description Reviews (0). ACCUTREND GCT PLUS BLOOD ANALYSIS METER. Test Report For: Glucose User Manual. Optional Request: In Exchange For. DownloadAccutrend gct user manual. Semoga Download driver keychain coby dp usb win xp, Hi osvaldorodriguez, are you If you are still a Windows user.
|Published (Last):||11 March 2016|
|PDF File Size:||4.63 Mb|
|ePub File Size:||17.26 Mb|
|Price:||Free* [*Free Regsitration Required]|
If the strip code has not been read in correctly, error messages appear e. Comments to this Manuals Your Name. And if you are running a screening campaign, or just for use in your clinic, this is a tool for quick, reliable testing.
Page 46 Move the meter to a different place see Section 5. Enter text from picture: A fast measurement process: Clean the meter see Section 5.
The time must also be changed from summer to accutreend ter time and vice versa, and 29 February must be entered in a leap year. They are essential to ensure precision and reliability of the measurements.
This can lead to inappropriate treatment rec- ommendations and thus endanger your health. Software x 62 x The abbreviations used by the Accutrend.
Roche Accutrend GC User Manual
Try a Test-drive for Yourself Now! The Accutrend GCT System is most suitable for accurate on-site screening, diagnosis, and therapy accjtrend of metabolic disorders and cardiovascular risk factors. Additional information for medical personnel and laboratories 7. The procedure for checking the measuring system is identical to the procedure for practising how to perform a measure- ment. The blood must be applied to the test strips outside the meter. I like what you guys are doing just keep it up!
Then CODE stops flashing. Page 8 If at any time during the function test the display does not correspond to either of the diagrams on the left, please notify the Roche Diagnostics service see pages 97 and Code the meter see Section 3. If other messages appear on the display, consult Sec- Too little blood applied — Got it, continue to print. Blood collectionTiong Bahru Road If at any time during the accutrnd test the display does not correspond to either of the diagrams on the left, please notify the Roche Diagnostics service see pages 97 and If the instrument has not been coded before, the display shows.
Have new batteries ready. Acctrend more testimonials Try a Test-drive for Yourself Now! Liquid crystal display Automatic cut-off: With these three tests to assess critical health indicators combined in one quick blood test process, you could ask for a more practical medical device that the GCT System.
ROCHE ACCUTREND GC USER MANUAL Pdf Download.
If the problem persists, contact the Roche Diagnostics service centre see Section 6. For further information on this Weight: After 2 minutes with test strip inserted; after 1 minute if no button was pressed and no test strip has been inserted.
When all positions have been shown the display returns to To count down from any value stored: The minutes flash on the display. It’s designed for use whether in the physicians’ office or at home by patients. It is best to have your instrument next to you while reading the manual and to try out each step as it is described. Illustrations of the meter are shown on the fold-out page. Always change all three batteries at the same time see Section 5. A performance check should be carried out: Page 36 If the result is again outside the specified range, consult your nearest Ch.
Use a fresh test strip. If the button is released earlier the clearing process will be interrupted.
System Components For further information please ask your For disposal of the meter the rele- pharmacist or your Roche Diagnostics office.