Fluke A Multimeter DIGITAL | Business & Industrial, Test, Measurement & Inspection, Test Meters & Detectors | eBay!. The listing you’re looking for has ended. Fluke A Multimeter. Condition: For parts or not working. Ended: Jun 08, , AM PDT. Price: US $ Corporation (Fluke) has reproduced this copy of a manual which is no All shipments of JOHN FLUKE MFG. CO. . A A rechargeable Ni Cad battery pack.
|Published (Last):||11 November 2011|
|PDF File Size:||18.25 Mb|
|ePub File Size:||7.25 Mb|
|Price:||Free* [*Free Regsitration Required]|
It is not intended to be used as a verification of calibration accuracy. Mouse over to Zoom – Click to enlarge. Use the following procedure for operating the A with the transformer: The value of Vref fluuke determined by algebraically adding Vx a positive voltage to a negative equivalent of the ohms source voltage. To find out more, including how flyke control cookies, see here: By putting two transistors in series we obtain two times the log of the input; then by taking the antilog we obtain a voltage proportional to the square of the input Vi 2.
Sign in to check out Check out as guest.
John Fluke Model 8040A Digital Multimeter
Input Divider See terms – opens in a new window or tab. Inattention to these precautions may lead to instrument damage.
The peg, when the battery pack is in place on the A, closes the charging path allowing the batteries to be recharged. Perform the A calibration as follows: The final step in determining the rms value is handled by U 1 3 pins 7, 6, flkue 5the lower right transistor of array Ull and the output filter the 47k resistor and 0. The number in each decade counter is converted to a four-bit binary code.
Parts lists include the following information: Similarly, a steady-state error or gradient exists between the measured surface and the sensing device in the probe tip.
The cover does not include the battery retainers used to secure the rechargeable Ni-Cad batteries to the cover. Delivery time is estimated using our proprietary method which is based on the buyer’s proximity to the item location, the shipping service selected, the seller’s shipping history, and other factors. Adjust the variable resistor on the Model 80F-5 for a 1. NOTE If the amplitude flke the first ms of either oscilloscope display significantly deviates from waveform A1 or Bl, the malfunction is probably in the Buffer circuit Ql through Q8, Qll, U5 and U4 or the associated circuitry.
This fuse is located behind the mA input jack and is removed by pressing in lightly on the jack then turning it counterclockwise! It is recommended that the assembly and initial operation of the A be done in accordance with the following procedure.
This includes items that are defective in ways that render them difficult to use, items that require service or repair, or items missing essential components. In the AC and DC functions the 8040x voltage is fixed at 1. Each power configuration has a unique model number which must be used when ordering this accessory.
This amount is subject to change until you make payment. This item will be shipped through the Global Shipping Program and includes international tracking. No additional import charges at delivery! U14 pin 3, VDC function, 2 range, 2. The oscilloscope display should be similar to waveform Bl.
The A display will indicate the battery voltage; between 4. Check the hoodie, which can be pulled out, in case you are doing measurements in bright light or outdoors, it will help you to see the display properly. The inputs to be measured are connected to the Analog circuitry via the V-H or mA terminals.
Full text of “FLUKE a”
Start of add to list layer. Learn More – opens in a new window or tab International shipping and import charges paid to Pitney Bowes Inc. Clean the front panel and case with denatured alcohol or a mild solution of detergent and water. See Table for recommended calibration equipment. The strobe signal causes the digit in the display corresponding to the digit with the same significant value transmitted from the Timing and Control circuit to appear in the correct position.
Add to watch list.
Fluke 8040A Handheld “Bench” Multimeter Repair/Restore
These alkaline batteries will typically provide 10 hours of operation before they need replacing. AC voltage and current measure- ments are made using true rms conversion techniques for improved accuracy in communications and industrial control applications. Three versions of the Model 80F are available.
The oscilloscope should display waveform A. A high voltage probe facilities connection to the measurement source. Have one to sell?