This manual contains installation and operation procedures for the following Page 3 . Considerations for Safe Operation of Your Biodex Multi-Joint System. Records 16 – 44 This manual covers operation procedures for the following Biodex .. Biodex Multi-Joint System 3, Rev 2 or higher, with the proper hardware. Bi doex. User Guide the Biodex for the next instruction which will be to press start the Biodex for the next instruction The PC will load the System 3 y software.
|Published (Last):||14 July 2009|
|PDF File Size:||3.32 Mb|
|ePub File Size:||8.53 Mb|
|Price:||Free* [*Free Regsitration Required]|
Rotate chair to 15 degrees on opposite side. Before pressing the Start button, always inform the subject that the sytem arm will now be able to move.
System 4 Software Update v – Software Updates – Physical Medicine | Biodex
Set dynamometer tilt to degrees9. Knee Attachments Left and Right Patterns: Press the Start key on the control panel. Adjust attachment length so that pad is positioned just superior to the popliteal fossa. SpeedUse the Speed Away and Toward buttons to set the maximum velocity for each direction of movement inthe Isokinetic, Passive or Reactive Eccentric mode.
Failure to do so may result in lost or damaged files. Always set new limits when testing a new subject or moving from one joint to the next. Pull thelever and hold it back while you rotate the footplate until the desired color-coded position tagaligns with the lever.
Overflow has been found to be plus or minus as giodex as 15 degrees. The dynamometer shaft will now begin to rotate and the system will display a message that it is initializing the dynamometer. Included is information on both mechanical and anatomical aspects. Relaxation can some- times be assisted by the application of heat, cold, or biofeedback. Torque limits can be set as low as.
Footplate Rotation Lever B. Rotate the dynamometer to the blue color codepositions for right side testing or exercise.
If you have any questions, contact Biodex service at in NY Off axis through the glenohumeral joint. Adapter Locking Knob F. These buttons are used in Setup mode, per patient protocol, to set the final safe or target maximum range biofex motion for the specific test or exercise to be performed.
Please do not attempt installation or repair on your own. Cervical Support Adjustment Knob6.
Slowly syxtem with caution, the percent ROM dials should be turned up. Install limb support angled toward patient in chair side receiving tube for side to be tested or exercised.
Have patient face opposite way and grasp handgrip. There are a number of bursae at the hip.
BIODEX SYSTEM 3 PRO MANUAL ( # 8 3 5 – 0 0 0 ) Pages 51 – – Text Version | FlipHTML5
Move patient into position and align axis of rotation. NEW Upper extremity biodwx for hemiparetic patients. Any sequence of modes can be employed automatically. In the Table of Contents, you More information.
System 4 Software Update v4.60
These guide-lines are of a mechanical nature and do not reflect use of the computer software. Isotonic Mode The Isotonic mode allows velocity to vary while providing inertia-free constant force and concentric or eccentric muscular contractions. Releasing the button locks the shaft at the new position. The eccentric mode may be used to strengthen the musculotendinous junction. Be sure to correctly align suggested anatomical axis with the dynamometer shaft.
Versatility and marketability System 4 is dependable and versatile — providing the means to take on new challenges and be adaptable to keep pace with progressive thinking and innovation. Subjects with impingement syndrome may best be worked in the modified neutral position and not 90 degrees of abduction.
Raise dynamometer to align axis of rotation. For additional information on use and care of the Closed Chain Attachment, refer to the ClosedChain Attachment Operations Manual which was supplied with this unit. During exercise, limits can be moved within, but never beyond, the ROM established in the Setup bipdex. The LED on the control panel for the second direction should now stop flashing.
Rotate Wrist attachment to horizontal do not remove. Use of any attachment mahual than theis outside proper and intended use. Position pad anteriorly on the thigh and secure by wrapping the padded strap snuggly around the thigh.