The Pocket Colorimeter II™ has two manuals. The Basic and Full User Manuals can be found below. All Pocket Colorimeter™ II instruments. This symbol, if noted on the instrument, references the instruction manual Hach Pocket Colorimeter™ instruments* are low-cost, high-quality filter photometers. Consult HACH’s Pocket Colorimeter II brochure on ArchiExpo. Page: 1/4.
|Published (Last):||28 February 2014|
|PDF File Size:||14.73 Mb|
|ePub File Size:||20.72 Mb|
|Price:||Free* [*Free Regsitration Required]|
Repeat for all points to be deleted.
User-Entered Calibration, continued 4. Remove power from the Pocket Colorimeter instrument by removing one of its batteries to verify that it is or is not the source of the interference.
Page of 88 Go.
Order information – Hach-Lange POCKET Colorimeter II User Manual
Combined chlorine represents a combination of chlorine-containing compunds including but not limited to monochloramine, dichloramine, nitrogen trichloride, — At least two data pairs concentration and absorbance are required. Please contact the Hach Service Center serving your location.
Silicon photodiode Photometric precision: User-Entered Calibration, continued See Interferences on page 1— Instrument Operation, continued 2. Warranty In the event that a defect is discovered during the warranty period, Hach Company agrees that, at its option, it will repair or replace the defective product or refund the purchase price, excluding original shipping and handling charges.
Open the catalog to page 1. A flashing arrow indicates MENU the current range. The curve can extend from 0 to 2. A tough, impact-resistant shell protects the electronics and optics, while a lanyard keeps the cap attached to the colorimeter. Precautionary Labels Please pay particular attention to labels and tags attached to the instrument.
High range free chlorine determinations are subject to variable levels of interferences from monochloramine. Cover the blank with the instrument cap. Standard Calibration Adjust, continued 7.
The polarities are shown on the battery holder. To enter the S0 value, press. You can adjust your Community Subscriptions in Settings.
Hach Company – Pocket Colorimeter II Community, Manuals and Specifications | MedWrench
Battery Installation Instrument Operation, continued Battery Installation Figure 1 on page 2—7 provides an exploded view of battery installation. Operation is subject to the following two conditions: Each data pair requires a concentration value and the absorbance value for the given concentration. For order information by E-mail: Open the catalog to page 2.
Accurate measurements cannot be obtained unless the sample or blank is covered with the cap.
Recalling Stored Measurements 1. Use of the factory calibration hafh recommended unless the user is required to generate a calibration. Insert the reagent blank or deionized water into the meter and cover with the cap.
Hach Pocket Colorimeter™ II for Chlorine or Manganese
Two options are available on the Pocket Colorimeter II to meet this requirement. Many are based on EPA approved methods. Both forms can coexist in the same solution and can be determined together as total chlorine. Printed in the U.
Page 79 Certification, continued 1. The Pocket Colorimeter II is simple enough to use that anyone can obtain reliable results with accuracy comparable to that of a lab instrument. Press the key twice to exit and accept the changes.
Order information | Hach-Lange POCKET Colorimeter II User Manual | Page 4 / 4 | Original mode
Enter text from picture: Repeat this sequence until the S0 concentration value is entered. Error Codes Error Codes When the instrument cannot perform the function initiated by the operator, an error message will appear in the display.
User-Entered Calibration, continued Note: In the event that a defect is discovered during the warranty period, Hach Company agrees that, at its option, it will repair or replace the defective product or refund the purchase price, excluding original shipping and handling charges.